Articles in this category Is LINEX available for PAL3 (or Shimadzu AOC6000)? Yes, we have LINEX-2 for the PAL3. The LINEX is fully integrated in the PAL3 starting from PAL firmware version 2.3. The follow Can I use 1/4 inch x 3.5 inch TD tubes with the LINEX? Yes it is possible to use LINEX with 1/4 inch TD tubes, but you need a special OPTIC versions for this. You need for the OPTIC: The liner drops out of the gripper. - If liners are dropped out of the gripper it is most of the times related to alignment. Make sure that the alignment is done correctly. - Also be sure Leak at the head of the LINEX / head does not close fully. Never test a leak with a soap solution like Snoopy. Always use an electronic leak detector like our LD239. Check also Can I use all methods in Evolution Workstation with the LINEX? No, only the LINEX methods and the EXPERT method can be used in combination with LINEX.These methods are special methods needed to make a good working system. The methods will initialise all setpoints... How can I work with an internal standard and TD tubes? When using LINEX for Thermal Desorption it is possible to introduce an internal standard in an automated way. The PAL System can inject an internal standard on the TD tube, this can be done both in t... What is the difference between; 'LINEX-Liquid Injection' and 'LINEX' method? LINEX Method: The LINEX method in Evolution Workstation is a special method for the liner exchanger. In a normal situation there is a pressure in the inl LINEX Method doesn't start when I use the Run Button or Run Method Icon The LINEX Methods in Evolution Workstation are methods made for the liner exchanger. This method expects a Run Signal from the PAL System and not from the operator. The 'Run What syringe can be used in the PAL3 LINEX-2 Tool? The LINEX-2 Tool for PAL3 is used to transport liners but it can also be used for liquid injections. The needle length should be 85mm and the glass ø6.6mm. CTC Tool compatibility = D7/85. Can the LINEX-2 for PAL3 work with the CTC PAL Sample Control software? Yes, the LINEX-2 and CDC station can both work in combination with the PAL Sample Control software. You need a set of Plugin's and example methods and sequences. This is all provided with a new LINEX... Hissing sound in the LINEX box / the LINEX head is not closing well. We have see with older systems that there can be a hissing sound inside the LINEX box and the head is not closing like before. Also this problem is sometimes not constant. We have found that the cylin... How should I condition a TD tube? Before TD tubes can be used the desorption tubes must be conditioned. Conditioning involves the continuous flow of a high purity carrier gas (either Nitrogen or Helium) while the desorption tubes are ... LINEX used with Chronos / PAL Sample Control ; setting of Overlapped When LINEX is used the overlapped setting should be Off. When set to On the methods will not work well. LINEX used with Chronos / PAL Sample Control ; setting of Overlapped When LINEX is used the overlapped setting should be Off. When set to On the methods will not work well.